An Introduction to Phybrata
There is an ever-growing body of work which aims to draw attention to the understanding that professional & amateur athletes competing, at all levels, in high-impact sports are more susceptible to long term brain damage. First coined as the ‘punch drunk syndrome’ in professional boxing, the neurodegenerative pathology, linked to exposure to repetitive, mild traumatic brain injury, or to a single, moderate or severe traumatic brain injury, is now known as Chronic Traumatic Encephalopathy (CTE) Stewart et al (2016), Stewart et al (2019).
In an attempt to address these concerns, sports where competitors may be subject to concussions (including, but not limited to, association football, American football, rugby, cycling, cricket & boxing) have all implemented a ‘Concussion Protocol’ designed to be adhered to following any incident which has potential to cause head injury. However, these protocols, are not standardised, the governing bodies who implement them do so unevenly and inconsistently around the the world, and it is often left to the discretion of a referee, trainer, coach or team doctor to determine whether or not a participant may continue.
Diagnosing concussion is often a difficult affair, given the multiple interacting physiological systems which can be impaired as a result of the injury. An inhibition of either the central nervous system, peripheral nervous system, sensory, neurovascular and muskuloskeletal systems could all demonstrate a concussion diagnosis. Whilst a variety of instrumented tools are available for the assessment of any of the aforementioned systems, it is a time consuming, costly endeavour, resulting in significant challenges for the prediction, diagnosis, treatment and rehabilitation of concussion. (Ralston et al (2020)) Despite worldwide acknowledgement, there remains no ‘gold standard’ assessment tool for the diagnosis of a concussion - it remains a clinical decision.
What is Phybrata?
Physiological vibration acceleration (phybrata) sensing is an innovative new approach to precision health monitoring pioneered by the American company, PROTXX. The standard phybrata test developed by PROTXX requires the patient to simply stand still for 20 seconds with their eyes open (Eo) and again for 20 seconds with their eyes closed (Ec). During this static balance testing, the phybrata sensor uses a microelectromechanical system (MEMS)-based inertial motion unit (IMU) to detect microscopic involuntary vibrational motions of the body (linear accelerations in 3 directions). The phybrata signals record both normal motions characteristic of healthy individuals and pathological motions caused by physiological impairments that degrade balance and postural stability. Figure 1 shows sample phybrata times series data and spatial scatter plots, as reported by the PROTXX iOS app, illustrating the visually observable differences between Eo vs. Ec tests, as well as between “healthy” and “impaired” test subjects. In this example the impairment was caused by a concussion injury.
Figure 1. Sample eyes open (Eo) and eyes closed (Ec) x(anterior-posterior, AP), y(vertical), z(medial-lateral, ML) acceleration time series data and AP/ML acceleration spatial scatter plots for age/gender-matched (A) healthy baseline participant (B) patient with diagnosed concussion.
PROTXX combines unique spatial-domain, time-domain and frequency-domain features in the phybrata signals into vibrational signatures (or digital biomarkers) which are used to identify and quantify impairments to specific physiological systems, including the central nervous system (CNS), peripheral nervous systems (PNS), sensory systems (visual, vestibular, somatosensory), and musculoskeletal system, as illustrated in Figure 2 below.
Figure 2. Phybrata signals are used to detect and quantify impairments to different physiological systems in the body based on their unique contributions to biomechanical stabilisation of the head and eyes as the body's reference platform for balance and movement.
For “impaired” individuals, additional phybrata biomarkers have been developed by PROTXX which allow impairments to the CNS, PNS, sensory (vestibular, visual, somatosensory), and musculoskeletal systems to be identified and quantified: For the specific case of concussions, the ROC curves in Ralston et al (2020) indicate that the average power (Eo+Ec)/2 may be utilised to support clinical diagnosis of concussion, while Eo and Ec/Eo may be utilised as independent measures to confirm accompanying neurological and vestibular impairments, respectively. All 3 measures demonstrate area under the curve (AUC), sensitivity, and specificity above 90% for their respective diagnoses. Note that the “cutoff” values for each metric generated by the receiver operating characteristic (ROC) analysis in the above reference and summarised in Table 4 (reproduced below) are the 95% confidence level threshold values for each metric to classify the presence of the corresponding impairment Ralston et al (2019).
The above results for concussion impairments match the diagnostic performance (sensitivity, specificity, accuracy, Area Under Curve (AUC)) of current clinical solutions that combine data from multiple tests (computerised dynamic posturography, eye tracking, neurocognitive assessments) to generate more complex multimodal concussion biomarkers. (Ralston et al (2020))
Figure 5 (below) shows sample Eo and Ec phybrata power histograms and Anterior-Posterior/Medial Lateral acceleration spatial scatter plots for 15 athletes (ages 17–23): 5 healthy athletes during baseline testing (Figure 5A); 5 following a concussion with only vestibular impairments observed during physical examination (Figure 5B); and 5 following a concussion with other neurological impairments observed during physical examination (Figure 5C). Patients with vestibular impairment only consistently showed Eo phybrata powers that remained below the 95% confidence interval (CI) for healthy subjects, but Ec phybrata powers elevated above the 95% CI for healthy subjects (Figure 5B). Patients diagnosed with neurological impairment, on the other hand, typically showed both Eo and Ec phybrata powers significantly elevated above the respective 95% CIs for healthy subjects (Figure 5C).
Immediately following each Phybrata Eo-Ec test, the Protxx-Clinic mobile app presents the patient’s phybrata performance along with the normative ranges expected for healthy individuals of the same age, allowing classification of the patient’s performance as “healthy” vs. “impaired”. Biomarkers for specific impairments of interest are also plotted together the 95% confidence intervals used to quantify the degree of impairment. The patient’s test results can also be compared to previous phybrata tests, and changes in the patient’s phybrata biomarkers as a function of time can be used to quantify and track the response of specific physiological impairments to medications, treatment, or rehabilitation. Having completed two stationary 20 second tests (1x eyes open, 1x eyes closed), the collected data is streamed to the cloud for signal processing, data analysis and report generation. Results of phybrata testing are made available in a number of clear, intuitive graphs (Figure 6). The first and second screens display visit notes and a patient identifier, with a summary of the results. The third screen of the report plots the patient’s Eo and Ec phybrata power, aligned to their age, on a graph of the average Eo and Ec powers of healthy individuals across a range of ages. This chart allows the subject and operator to see how the results from the current test relate to the average values for healthy people the same age.
Three phybrata measures are derived from the raw sensor data:1) Phybrata power
2) Eyes-Closed/Eyes-Open Phybrata power ratio (Ec-Eo Ratio)
3) Left/Right Phybrata power ratio (L/R Asymmetry)
Figure 6: Test Report Screens - a) Visit Notes, b) Report Summary, c) Age Comparison with Healthy Subjects, d) Phybrata Power & L/R Asymmetry Bar Charts
The Acceleration Scatterplot screen (Figure 7) displays spatial scatter plots that illustrate the patient’s movement during both the Eo and Ec tests. The scatter plots are presented from a top-down viewpoint, with axis labels for Forward-Backward and Left-Right movement. The dots represent accelerations of the head at each point in time during the test, measured in G’s. Typically a good phybrata test will have tight, symmetrical scatter plots with all the accelerations densely focused around the centre point. Ec phybrata plots typically show observably greater scatter than Eo phybrata plots.
Figure 7. Test report screens. e) Acceleration Raw Data Traces, f) Acceleraton Scatter Plots
The Acceleration Raw Data Trace screen displays the raw accelerometer traces for both Eo and Ec tests. These traces represent the accelerations of the head, measured in G’s, as a function of time during the 20 sec Eo and Ec tests. The red trace represents Forward-Backward accelerations of the head, where forward is (+) and backward is (-). - The green trace represents Up-Down accelerations of the head, where up is (+) and down is (-). - The blue trace represents Left-Right accelerations of the head, where Left is (+) and Right is (-). Most motion will appear in the Back-Fwd (red) and Left-Right (blue) traces. Typical motion will have some longer movements from side to side as well as some shorter ones. The best indicator of good health is that all three traces remain relatively close to 0. You should see some movement, especially in the Forward-Backward (red) and Left-Right (blue) traces, but no abrupt spikes or large deviations from zero.
This phybrata sensor, manufactured by PROTXX, available from Brainbox, provides a simple, wearable, non-invasive, and clinically intuitive alternative to more complex multi-systems approaches currently adopted for objective identification and quantification of impairments and sensory reweighting affecting multiple physiological systems following concussion injuries. Phybrata testing contributes to a more comprehensive picture of each patient’s unique impairment signature and helps guide targeted rehabilitation strategies and track recovery trajectories, both in the clinic and via remote patient monitoring. The results present a consistent and cohesive pattern that integrates complimentary results from current research and practice in industrial, athletic, and clinical medicine, and highlight significant opportunities to improve clinical assessments, recovery management, and return-to-activity decision-making for concussion patients. (Ralston et al (2020))
Chronic Traumatic Encephalopathy: The Neuropathological Legacy of Traumatic Brain Injury. Jennifer Hay, Victoria E. Johnson, Douglas H. Smith & William Stewart. Annual Review of Patholoy: Mechanisms of disease. January 2016
Chronic traumatic encephalopathy is a common co-morbidity, but less frequent primary dementia in former soccer and rugby players. Edward B. Lee, Kevin Kinch, Victoria E. Johnson, John Q. Trojanowski, Douglas H. Smith & William Stewart. Acta Neuropathalogica. June 2019.
Physiological Vibration Acceleratuon (Phybrata) sensor assessment of multi-system physiological impairments and sensory reweighting following concussion. Ralston JD, Raina A, Benson BW, Peters RM, Roper JM, Ralston AB. Medical Devices: Evidence and Research. 2020
Monitoring of postural sway with a head-mounted wearable device: Effects of gender, participant state, and concussion. Grafton ST, Ralston AB, Ralston JD. Medical Devices: Evidence and Research. 2019
Brainbox Initiative Conference 2021
London, United Kingdom
23 Sep 2021
Virtual Exhibition Hall
11 Nov 2021
4th International Brain Stimulation Conference
Charleston, South Carolina, USA
6 Dec 2021
Breakfast Workshop: Introduction to Focused Ultrasound Neuromodulation
Charleston, South Carolina
8 Dec 2021